measured when probes are open circuit is not close to zero
variations in open circuit capacitance are normal, perform
a probe compensation procedure.
and/or inductance measured when probes are short circuit
is not close to zero (±1.2 ohms, ±1.6uH)
variations in short circuit resistance and inductance are
normal, perform a probe compensation procedure.
value doesn't appear to be correct.
probes are well connected to the component under test for
the entire duration of the analysis.
Ensure that nothing else is
connected with the component under test. Make sure you are
not touching the connections.
The component value may be outside
the supported measurement range.
The component's design frequency may
not correspond to the test frequencies used by the LCR45.
values seem to vary over time.
displayed resolution is finer than the internal
measurement resolution to avoid rounding errors. Small
variations within the quoted measurement resolutions are
date is approaching or has past.
worry, the LCR45 will carry on working after the
"Calibration Due Date" has past. The date is
simply a recommendation and may be required for the
purposes of satisfying your own quality procedures. Click
here if you would like us to recalibrate your unit.